1. Handle with care during use and do not collide with other hard objects; Avoid internal IC shedding and damage.
2. Pay attention to the cleaning of the KIC tester during use, and avoid some sundries falling into the IC slot, resulting in poor IC contact.
3, in the furnace must be covered with heat insulation performance good 1/4inch thick heat insulation sleeve, avoid in the furnace when high temperature broiling instrument; And often high temperature baking, shortened the service life of KIC.
4, when using the battery, must use alkaline battery, avoid in the high temperature test process voltage instability, resulting in incomplete curve test. When replacing the battery, pay attention to the good contact between the battery head and the battery.
5, when using, please follow the KIC standard test steps (first communicate with the computer -> and then pass the furnace test -> and finally download the curve) to test.
6. During the next temperature curve test, ensure that the internal temperature of the KIC tester is lower than 35 degrees, so as to avoid damaging the instrument due to the increase of internal temperature caused by repeated tests.
7, in the plug and pull thermocouple, do not two or more at the same time to plug and pull, this will easily cause damage to the KIC furnace temperature tester channel.
8. In use, please ensure that AirTC solder joint is in good condition, because it not only controls the beginning and end of recording temperature data of the instrument, but also provides data to the instrument for calculation of thermal efficiency. Bad contact of AirTC will directly affect data loss and incomplete.